No Supervision by ATC, CFS, or Anyone of Annual/Periodic Inspection SOPs
To show that the ATC recommendations for operational procedures at time of use do not account for the need during inspections to regularly conduct control tests at target values other than 100 mg/100mls.
To demonstrate that we have no metrological supervision to ensure that linearity is checked, that manufacturer's specifications are honoured across the measuring interval.
To demonstrate that searches of breath using"approved instruments" are unreasonable in Canada if there is no system in place to supervise and audit police maintenance of their instruments.
To explore lack of metrological supervision of police and authorized service centres respecting inspection of approved instruments to ensure that they continue to conform to manufacturer's specifications